These tutorial movies show the installation and qualification of a DECADE Elite electrochemical detector and flow cell. The procedure is also described in the manuals in the download section for DECADE Elite, SenCell and FlexCell. Preparation of a flow cell before use is on another page.
|STEP 1. Unpacking and powering up. Set the temperature to stabilize in preparation of the OQ test
|STEP 2. Install the dummy cell and let it stabilize in preparation of the OQ test. In total, about one hour stabilization time is needed before starting the OQ (step5).
|STEP 3. Connect the cables for data acquisition and autozero. This example shows a special Shimadzu AD cable.
Skip this step: in case Clarity CDS or another CDS with Antec ECD control driver is used.
|STEP 4A. Install Dialogue Elite software in preparation of the OQ test.|
|STEP 4B Clarity installation|
STEP 4C. Install PCI-serial card drivers
|In case the PC is equipped with Antec's PCI - serial card, it is now time to install the drivers. Click here to download the zip file and install the drivers.|
|STEP 5. Perform the OQ test – Part 1: Dummy cell noise test|
|STEP 6. Perform the OQ test – Part 2: Analogue output test|
STEP 7. Familiarize an end-user with the instrument and explain how to do maintenance and troubleshooting. A set of useful videos for this purpose can be found here.
STEP 8. Complete and sign the IQOQ document after finishing.