DECADE Elite installation – steps
DECADE Elite and Lite.
This video show the steps to a successful installation of the DECADE Elite. Our tutorial videos are available for visual support of the operational procedures. It does not replace the written documentation, always first read instructions and safety precautions in the manuals of related instruments and cells.
Unpacking and powering up. Set the temperature to stabilize in preparation of the OQ test
Install the dummy cell and let it stabilize in preparation of the OQ test. In total, about one hour stabilization time is needed before starting the OQ (step5).
Connect the cables for data acquisition and autozero. This video shows a special Shimadzu AD cable (L) or another brand (R). Skip this step: in case Clarity CDS or another CDS with Antec ECD control driver is used.
In case it is part of the system, install Clarity and if needed, also Dialogue Elite software.
Install PCI-serial card drivers. In case the PC is equipped with Antec’s PCI – serial card, it is now time to install the drivers. Click here to download the zip file and install the drivers.
Perform the OQ test – Part 1: Dummy cell noise test. Part 2: Analogue output test
Familiarize an end-user with the instrument and explain how to do maintenance and troubleshooting. A set of useful videos for this purpose can be found here. Complete and sign the IQOQ document after finishing.
Change PC IP address
The IP address can be changed in the Windows 10 control panel. See also our KB paper on this topic.