No peaks, no sample is introduced
HPLC with autosampler
No, or too small signal/front peaks visible in chromatogram.
Blockage of AS flow path or leakage in the injection valve.
Diagnosis and Remedy
Follow the trouble-shoot instructions in the flow diagram below. See also: ‘no signal‘ for detector related issues.