Welcome to our Knowledge Base
Continuous overload of cell current
Signal is continuously in overload.
- Incompatible detector settings: a high cell potential in combination with a sensitive range setting.
- Some other problem in LC or ECD: see high background current.
- Switch to a less sensitive range setting, check if less extreme potential can be used.
- For a problem in LC or ECD: see high background current.