OQ PQ PM cal hardware kit (pn. 250.3060) contains among others, the PQ FIA tubing set (left), and the dedicated cables and plugs (right) to perform OQ, PQ and PM calibration procedures.
Antec developed procedures and kits for Installation Qualification (IQ), Operational Qualification (OQ), Performance Qualification (PQ) and Preventive Maintenance Calibration (PM-Cal) for Antec detectors, ALEXYS and ROXY systems. Each procedure is described in a document that includes a sign-off section for the engineer and customer.
For starters, the ‘OQ PQ PM cal hardware kit’ (pn. 250.3060) is available. This kit covers all the reusable cables, plugs and the ‘professional version’ software dongle of the Dialogue Elite software, which are required for a service engineer to be able to perform Antec’s Qualification Procedures. The individual parts needed per device/procedure are conveniently listed in the table below in cases where a partial kit needs to be supplemented. In addition to the listed parts, the service engineer should bring their own voltmeter and temperature probe, and a consumables kit for PQ as specified below.
Our Dialogue software (professional version) applies a convenient and user-friendly test script that generates a standardized Excel file as output for the OQ and PM-Cal procedures.
|171.9012||1||Dialogue Elite professional|
|250.0040||2||External dummy flow cell|
|250.0128B||1||DECADE Elite DCC output cable D9 - banana
|250.0150||1||Ecell test cable D conn.|
|250.0152||1||DECADE II output cable BNC to banana|
|250.0153||2||10 KOhm OQ resistor|
|250.0154||2||1.0 MOhm OQ resistor|
|250.0156||1||I/O test connector for DECADE II control board|
|250.0158||2||I/O test connector for DECADE II sensor board|
|250.0159||1||I/O test connector for DECADE Elite|
|250.0175||1||USB cable A-B, 3m|
|250.1052||2||PQ FIA tubing set|
The table below indicates which parts is necessary for which procedure and instrument
|10 KOhm OQ resistor||250.0153||+||+||-||-||-|
|1.0 MOhm OQ resistor||250.0154||+||+||-||-||-|
|Thermometer for calibration||250.3056||+||+||-||-||-|
|Ecell test cable D conn.||250.0150||+||+||-||-||-|
|I/O test connector for DECADE II control board||250.0156||+||-||-||-||-|
|I/O test connector for DECADE II sensor board||250.0158||+||-||-||-||-|
|I/O test connector for DECADE Elite||250.0159||-||+||-||-||-|
|DECADE II output cable BNC to banana||250.0152||+||-||+||-||-|
|DECADE Elite DCC output cable D9 - banana||250.0128B||-||+||-||+||-|
|Dialogue Elite professional||171.9012||+||+
|External dummy flow cell||250.0040||+||+||+||+||-|
|PQ FIA tubing set||250.1052||-||-||-||-||+|
|PQ consumables kit DC mode on GC||250.3048||-||-||-||-||+|
|PQ consumables kit PAD mode on Au||250.3046||-||-||-||-||+|
* An OQ is part of the complete PQ procedure. Therefore, the OQ parts are also necessary when performing the PQ
** Alternative to the voltmeter is an AD convertor for the OQ test
Contact us for a quote or any other questions.